The most likely suspect would be loose connectors on the back of cluster. Download 95 service manual @ following link. It will instruct you on how to take cluster out. My Files
Another probability is the printed circuit on back of cluster is going bad. Or the nutd attaching leads of instruments are not making good contact & need tightening.